# changelog

author: Richard Hughes <rhughes@redhat.com> - 1.4.0-6
changelog:
  - Drop 55-Argyll.rules, it's not required and we can rely on colord
    to provide the ENV{COLOR_MEASUREMENT_DEVICE}="1" without the
    plugdev group or invoking a usb-db instance for each USB device
    hotplug.


author: Richard Hughes <rhughes@redhat.com> - 1.6.2-1
changelog:
  - Update to 1.6.2
  - Added "dark region emphasis" -V parameter to targen and colprof
  - Changed i1d3 driver to be more forgiving of EEProm checksum calculation
  - Fixed "edges don't match" bug in printarg when -iCM -h -s/-S used.
  - Fixed bug in -H flag in chartread, dispcal, dispread, illumread & spotread
  - Fixed bug in dispcal black point optimization to err on the black side
  - Fixed bug introduced into ColorMunki (spectro) reflective measurement
  - Fixed major bug in illumread - result was being corrupted.
  - Fixed problem with TV encoded output and dispread -E -k/-K


author: Richard Hughes <rhughes@redhat.com> - 1.6.3-1
changelog:
  - Update to 1.6.3
  - Added code to minimize ICC rounding error white point accuracy
  - Changed colprof to deal with variable grid distribution in a more neuanced way
  - Changed colprof to used a power_like function for the grid distribution shape
  - Changed i1d3 driver to completely ignore any EEPROM checksum errors
  - Fix bug in xicclu -py conversion.
  - Fixed bug in dispcal to use the final measurement pass for the calibration
  - Fixed bug in spotread, dispcal & dispread for CCSS capable instruments
  - Renamed verify to colverify to avoid clash with MSWin program of the same name
  - Switch dispread to use NoClamp readings
  - Tweaked dispcal to try and improve accuracy of black point calibration


